Download HP A RF IMPEDANCE-ANALYZER OPERATION SM service manual & repair info for electronics experts. The HP A RF Impedance Analyzer measures 14 parameters Parameter measured: IZI, Swept measurement: manual or automatic sweep from start to. Agilent A Test Fixture. Operation and Service Manual. Third Edition. Agilent Part No. February Printed in Japan.
|Published (Last):||16 June 2004|
|PDF File Size:||10.97 Mb|
|ePub File Size:||19.2 Mb|
|Price:||Free* [*Free Regsitration Required]|
IIRC, the sweep parameters were something like this: Sorry, that last line of code there is supposed to read: It’s 84MB, so I hope you have a fast link. Guess you never know until you try! Manua, have worked with the A and, as I recall, the commands are nothing like any other instrument I have seen.
Message 5 of 6. Most Active Software Boards: In particular, after setting up and initiating the instrument, I am initiating the measurement and reading it as follows:.
HP A for Sale|Impedance Analyzers|Impedance Measurement Products|Test Equipment Center
Matt – you’re right, the commands for the A are completely different from the manusl probably because of its age. In the code above, MHz is an arbitrary frequency. Message 1 of 6. I suspect that this difference might be causing my problems. Accepted by topic author Matt C.
4191A RF IMPEDANCE ANL
I have not worked manjal that instrument, but the Agilent web site has the operating and service manual here. Thanks for your help guys. When I gave it a try, though, it worked fine.
However, rather than being returned the data at MHz, I am returned the data at whatever frequency the analyzer is currently set at. If not, what would creating a driver for it entail? Would you mind posting the code you used on the A?
I think the would probably take the old commands, but the ALSO has new commands. The forums emoticons took over for a bit there. Message 3 of 6. Auto-suggest helps you quickly narrow down your search results by suggesting possible matches as you type.
Keysight / Agilent A – In Stock, We Buy Sell Repair, Price Quote
Message 2 of 6. You really need the manual.
In particular, after setting up and initiating the instrument, I am initiating the measurement and reading it as follows: Message 6 of 6. I am able to communicate with the device through the GPIB interface, and am also able to retrieve information.
I went through the manual before, but when I saw their GPIB section, the strings were so convoluted and so unlike anything I’ve ever seen before that I assumed it was referring to something different. Good luck with it Should I be looking at doing it with LabView or something else? Message 4 of 6. Has anyone done anything on interfacing with the A since Matt C’s last post?